|Introduction to SQL|
|Form Query User Guide|
|Contact Help Desk|
Photometric Flux Calibration
The objective of the photometric calibration process is to tie the SDSS imaging data to an AB magnitude system, and specifically to the "natural system" of the 2.5m telescope defined by the photon-weighted effective wavelengths of each combination of SDSS filter, CCD response, telescope transmission, and atmospheric transmission at a reference airmass of 1.3 as measured at APO.
The calibration process ultimately involves combining data from three telescopes: the USNO 40-in on which our primary standards were first measured, the SDSS Photometric Telescope (or PT) , and the SDSS 2.5m telescope. At the beginning of the survey it was expected that there would be a single u'g'r'i'z' system. However, in the course of processing the SDSS data, the unpleasant discovery was made that the filters in the 2.5m telescope have significantly different effective wavelengths from the filters in the PT and at the USNO. These differences have been traced to the fact that the short-pass interference films on the 2.5-meter camera live in the same vacuum as the detectors, and the resulting dehydration of the films decreases their effective refractive index. This results in blueward shifts of the red edges of the filters by about 2.5 percent of the cutoff wavelength, and consequent shifts of the effective wavelengths of order half that. The USNO filters are in ambient air, and the hydration of the films exhibits small temperature shifts; the PT filters are kept in stable very dry air and are in a condition about halfway between ambient and the very stable vacuum state. The rather subtle differences between these systems are describable by simple linear transformations with small color terms for stars of not-too-extreme color, but of course cannot be so transformed for very cool objects or objects with complex spectra. Since standardization is done with stars, this is not a fundamental problem, once the transformations are well understood.
It is these subtle issues that gave rise to our somewhat awkward nomenclature for the different magnitude systems:
Previous reductions of the data, including that used in the EDR, were based on inconsistent photometric equations; this is why we referred to the 2.5m photometry with asterisks: u*g*r*i*z*. With the DR1, the photometric equations are properly self-consistent, and we can now remove the stars, and refer to u g r i z photometry with the 2.5m.
Overview of the Photometric Calibration in SDSS
The photometric calibration of the SDSS imaging data is a multi-step process, due to the fact that the images from the 2.5m telescope saturate at approximately r = 14, fainter than typical spectrophotometric standards, combined with the fact that observing efficiency would be greatly impacted if the 2.5m needed to interrupt its routine scanning in order to observe separate calibration fields.
The first step involved setting up a primary standard star network of 158 stars distributed around the Northern sky. These stars were selected from a variety of sources and span a range in color, airmass, and right ascension. They were observed repeatedly over a period of two years using the US Naval Observatory 40-in telescope located in Flagstaff, Arizona. These observations are tied to an absolute flux system by the single F0 subdwarf star BD+17_4708, whose absolute fluxes in SDSS filters are taken from Fukugita et al. 1996 As noted above, the photometric system defined by these stars is called the u'g'r'i'z' system. You can look at the table containing the calibrated magnitudes for these standard stars.
Most of these primary standards have brightnesses in the range r = 8 - 13, and would saturate the 2.5-meter telescope's imaging camera in normal operations. Therefore, a set of 1520 41.5x41.5 arcmin2 transfer fields, called secondary patches, have been positioned throughout the survey area. These secondary patches are observed with the PT; their size is set by the field of view of the PT camera. These secondary patches are grouped into sets of four. Each set spans the full set of 12 scan lines of a survey stripe along the width of the stripe, and the sets are spaced along the length of a stripe at roughly 15 degree intervals. The patches are observed by the PT in parallel with observations of the primary standards and processed using the Monitor Telescope Pipeline (mtpipe). The patches are first calibrated to the USNO 40-in u'g'r'i'z' system and then transformed to the 2.5m ugriz system; both initial calibration to the u'g'r'i'z' system and the transformation to the ugriz system occur within mtpipe. The ugriz-calibrated patches are then used to calibrate the 2.5-meter's imaging data via the Final Calibrations Pipeline (nfcalib).
Monitor Telescope Pipeline
The PT has two main functions: it measures the atmospheric extinction on each clear night based on observations of primary standards at a variety of airmasses, and it calibrates secondary patches in order to determine the photometric zeropoint of the 2.5m imaging scans. The extinction must be measured on each night the 2.5m is scanning, but the corresponding secondary patches can be observed on any photometric night, and need not be coincident with the image scans that they will calibrate.
The Monitor Telescope Pipeline (mtpipe), so called for historical reasons, processes the PT data. It performs three basic functions:
The Final Calibration Pipeline
The final calibration pipeline (nfcalib) works much like mtpipe, computing the transformation between psf photometry (or other photometry) as observed by the 2.5m telescope and the final SDSS photometric system. The pipeline matches stars between a camera column of 2.5m data and an overlapping secondary patch. Each camera column of 2.5m data is calibrated individually. There are of order 100 stars in each patch in the appropriate color and magnitude range in the overlap.
The transformation equations are a simplified form of those used by mtpipe.
Since mtpipe delivers patch stars already calibrated to the
2.5m ugriz system, the nfcalib transformation equations have the following
Assessment of Photometric Calibration
With Data Release 1 (DR1), we now routinely meet our requirements of photometric uniformity of 2% in r, g-r, and r-i and of 3% in u-g and i-z (rms).
This is a substantial improvement over the photometric uniformity achieved in the Early Data Release (EDR), where the corresponding values were approximately 5% in r, g-r, and r-i and 5% in u-g and i-z.
The improvements between the photometric calibration of the EDR and the DR1 can be traced primarily to the use of more robust and consistent photometric equations by mtpipe and nfcalib and to improvements to the PSF-fitting algorithm and flatfield methodology in the Photometric Pipeline (photo).
Note that this photometric uniformity is measured based upon relatively bright stars which are no redder than M0; hence, these measures do not include effects of the u band red leak (see caveats below) or the model magnitude bug.
How to go from Counts in the fpC file to Calibrated ugriz magnitudes?
Asinh and Pogson magnitudes
All calibrated magnitudes in the photometric catalogs are given not as conventional Pogson astronomical magnitudes, but as asinh magnitudes. We show how to obtain both kinds of magnitudes from observed count rates and vice versa. See further down for conversion of SDSS magnitudes to physical fluxes. For both kinds of magnitudes, there are two ways to obtain the zeropoint information for the conversion.
On a related note, in DR1 one can also use relations similar to the above to estimate the sky level in magnitudes per sq. arcsec (1 pixel = 0.396 arcsec). Either use the header keyword "sky" in the fpC files, or remember to first subtract "softbias" (= 1000) from the raw background counts in the fpC files. Note the sky level is also given in the tsField files. This note only applies to the DR1 and later data releases. Note also that the calibrated sky brightnesses reported in the tsField values have been corrected for atmospheric extinction.
Conversion from SDSS ugriz magnitudes to AB ugriz magnitudes
Due to a variety of factors, the SDSS ugriz magnitudes are not perfectly zeropointed onto an AB system as defined by Oke & Gunn 1983.
Generally, this is fine, but there are cases when one wants the best available estimate to an AB magnitude (e.g., fitting isochrones and galaxy spectral energy distribution templates).
As Section 4.5 of the EDR paper discusses, putting these magnitudes on a consistent AB system (i.e. converting the magnitude zeropoint into physical flux units; for the math, see converting magnitudes into fluxes below) is tricky. We expect that our u g r i z photometry is not exactly on the AB system, but we estimate from several lines of argument that the g-r, r-i, and i-z colors are nearly AB to within about 3%. We estimate that the u-g color is too red by about 5% (again with about 3% of uncertainty), in the sense that (u-g)(AB) = (u-g)(SDSS) - 0.05. We are continuing to work on measuring and cross-checking the exact photometric AB zeropoints.
Note that our relative photometry is quite a bit better than these numbers would imply; repeat observations show that our calibrations are better than 2%.
Conversion from SDSS ugriz magnitudes to physical fluxes
As explained in the preceding section, the SDSS system is nearly an AB system. Assuming you know the correction from SDSS zeropoints to AB zeropoints (see above), you can turn the AB magnitudes into a flux density using the AB zeropoint flux density. The AB system is defined such that every filter has a zero-point flux density of 3631 Jy (1 Jy = 1 Jansky = 10-26 W Hz-1 m-2 = 10-23 erg s-1 Hz-1 cm-2).
Then you need to apply the correction for the zeropoint offset between the SDSS system and the AB system. We do not know this correction yet, so the fluxes you obtain by assuming that SDSS = AB may be affected by a systematic shift of probably at most 10%.